WASHINGTON (AP) - Health products giant Johnson & Johnson on Monday issued yet another product recall, this one for OneTouch VerioIQ blood glucose meters sold in the U.S. and other countries.
They're being recalled because when a diabetic's blood sugar level is dangerously high, they do not provide a warning and instead turn off or they display an inaccurate reading.
The meters are made by J&J's LifeScan unit, which will issue a free replacement meter to all patients.
The world's biggest provider of health care products said Monday that the meters being recalled in the U.S. shut down when a patient's blood sugar hits 1,024 milligrams per deciliter. That's an extremely high level requiring immediate medical attention.
It's extremely unlikely that a diabetic's blood sugar level would get that high. However, if a patient experienced such extreme high blood sugar and did not get prompt treatment or got an incorrect treatment, that could result in a serious health risk or death, according to J&J. People with diabetes are encouraged to keep their peak blood sugar level, shortly after a meal, at or below 160 milligrams per deciliter.
When blood sugar stays even a little above that level, over time the excess sugar can damage blood vessels, the heart, kidneys, eyes and other organs, eventually resulting in severe complications such as blindness, kidney failure, leg amputations and premature death. Extremely high, sudden spikes can cause death if not treated aggressively.
Patients can continue using the meter until they receive a new one, the company said. Outside the U.S., the company is recalling the OneTouch VerioIQ, OneTouch VerioPro and OneTouch VerioPro+ Brands. The three types are being recalled because, at extremely high glucose levels, they display an incorrect glucose level or don't store the correct glucose level in their memory.
To date, no patient injuries related to the defects have been reported worldwide for any of the recalled meters.
The recall is J&J's latest in a string of about three dozen since 2009 by Johnson & Johnson. Most have involved nonprescription medicines such as adult and children's Tylenol and Motrin, but other recalls were for faulty hip replacements and prescription drugs for conditions such as epilepsy or for contact lenses. Reasons have included wrong levels of active ingredients in medicines, glass or metal shards in liquid medicines and nauseating packaging smells.
The company is operating under increased scrutiny from the U.S. Food and Drug Administration, while it completely rebuilds one nonprescription medicine factory from the ground up and upgrades other factories. The recalls and lost product sales have cost J&J well over $1 billion.
(Copyright 2013 The Associated Press. All rights reserved. This material may not be published, broadcast, rewritten or redistributed.)
Sunday, July 27 2014 11:15 PM EDT2014-07-28 03:15:51 GMT
Authorities say a father was killed and his daughter seriously injured while walking on the sand when a small plane crash landed along Florida's Gulf Coast near Venice Beach.More >>
A father was killed and his 9-year-old daughter critically injured Sunday while walking along a Florida beach when a small plane crash landed on the Gulf Coast only feet from the surf, authorities said.More >>
Monday, July 28 2014 1:54 AM EDT2014-07-28 05:54:28 GMT
Authorities now say 14 people - including a 15-year-old - have been struck by lightning during rare summer thunderstorms that have swept through Southern California, injuring two critically.More >>
A 20-year-old man died Sunday after lightning struck or rattled 13 people at a popular Los Angeles beach and a golfer on Catalina Island during summer thunderstorms that swept through Southern California, authorities said.More >>
725 S. Floyd Street
Louisville, KY 40203